Vag-Com measuring blocks channel 01 then 000
matt_20v at yahoo.com
Tue Apr 15 12:26:50 EDT 2003
I will check and document the exact procedures
and associated Bentley tables tonight, and post
here tomorrow. The terminology differences between
the Bentley and the tool are largely the root of
--- SuffolkD at aol.com wrote:
> Good to see you know what's what.
> I posted to Ross-tech for an explaination what they
> were but they're on
> are they the Bentley values on Table D2 - 440 -1 in
> Volume #1?
> Do you know how to assign values to Vag-scope?
> ......for Graphing?
> -Scott in BOSTON
> In a message dated 4/15/03 12:11:20 PM Eastern
> Daylight Time,
> 200q20v-request at audifans.com writes:
> > Anyhow, the ox sensor channel 8 is doing its thing
> > and hovering around 128 as it should. Channel 9
> > (ox sensor adaptation) is high- 160-ish vs. spec
> > 120-140. Bentley suggests this could be caused by
> > an intake air leak. I'll pursue that next oppty.
> > log some garage time. Getting closer to sorting
> > this new engine, and learning a few things in the
> > process.
> > Matt Rooke
> > '91cq//20vt
Do you Yahoo!?
The New Yahoo! Search - Faster. Easier. Bingo
More information about the 200q20v